- Last Active
- Registered Guest
So I changed the specs to read 3% of the gauge tolerance (10.185 - 10.815) instead of the ±2.5 of the part. The GRR is now a 74.094% and the NDC hasn't moved. Holding steady at .348. This has to be something with my resolution. Can someone please co…
Excellent News. Thank you so very much for the update. I will print these and get them to my manager.
Where can I find the PDF you talked about in part 3? How can I go about getting some reading material on the Gagepack. I use Gagetrak now and there's no support like what you have here.